Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
|Published (Last):||2 July 2012|
|PDF File Size:||6.4 Mb|
|ePub File Size:||2.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
The HP B is set at the factory for high impedance, refer to section 2 of the Operation and Calibration Manual if you 866b to set the input impedance to 50S1.
Performing this feature clears any previously selected function. In the Amplitude function, leading leros are not counted.
The Signal Generator frequency is h set to the reference frequency and amplitude to 0 dBm. Use a pointed object such as a pencil to set the switches to the desired HP-IB address in binary see Figure Ouput signal phase is adjustsdole in 1 degree nominal increments. Verify that the meastured Total Harmonic Distortion does not exceed 0.
Refer to page To avoid dangerous elec- tric shock, do not perform any seruicing unless qualified to do so. When you require best level accuracy. Verify that the measured frequency does not exceed the Umits specified.
Full text of “HP: B B final”
To clear this error condition enter a new carrier frequency in the correct hand for the selected Peak Deviation. Procedure Address Recognition This check determines whether or not the Signal Generator recognizes when it is being addressed and when it is not.
T Distortion Analyzer Hpp Range: The Rise and Fall Time should be within the limits shown in the Table.
Servicing instructions are for use by service trained personnel only. Hardware execution time will also vary considerably. Therefore, it is best to assume that the entire sequence of events shown in Figure occurs every time a 86656b change is made. The detailed operating instructions provide comprehensive information about all of np Signal Generator functions and features.
Ensure that the rating of the line mains fuse is appropriate for the line voltage being used, see TABLE Upon entering that value the measuring receiver will also change the center frequency t y that amount, the Signal Generator should be stepped to match. Frequency Amplitude Modulation 2.
HP 8656B Manuals
Unless otherwise stated the information applies to all three Signal Generators. Set the mmanual receiver with the sensor module precalibrated and connected as follows: If we find a better copy in the future, we will add it to the Agilent website.
Switch off the HP B and remove the power cable. The Signal Generator RMT and ADRS annunciators will both light, and its output signal and all preselected functions remain unchanged when the local-to-remote transition occurs.
Figure also shows the cable connections for this measurement. After you have connected the cables, press the Proceed key to nm the measurement. Step mmanual amplitude down to -2 dBm.
Hewlett-Packard B Signal Generator
Depending on the use and environmental conditions, the Signal Generator should be checked using the following performance tests at least once each year. Plug in the HP B and switch on. The Remote message is comprised of two parts. The 88656b is the same for all identical instruments. If the Signal Generator has been repaired or the heterodyne band of frequencies do not pass, the Optional Residual FM Test steps should be performed.
Directions to Display a Memory Location: Connect the channel 2 probe to A4TP12, 3. NOTE fVhen you select a timebase input or output frequency, the Signal Generator timebase output or timebase input, will be set for the same frequency. Model Frequency Result Min, Max.