Accutrend GC Blood Pressure Monitor pdf manual download. Since each code strip and test Glucose ® Accutrend, GC, GCT, DM CODE 50 TESTS. Description Reviews (0). ACCUTREND GCT PLUS BLOOD ANALYSIS METER. Test Report For: Glucose User Manual. Optional Request: In Exchange For. DownloadAccutrend gct user manual. Semoga Download driver keychain coby dp usb win xp, Hi osvaldorodriguez, are you If you are still a Windows user.
|Country:||Republic of Macedonia|
|Published (Last):||15 January 2006|
|PDF File Size:||6.75 Mb|
|ePub File Size:||20.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
On target high relevance: I collected it in good condition. If the result is again outside the specified range, consult your nearest Ch. But you surprised me and delivered on target. Users in the office and laboratory sector should note that samples containing human material must be handled as potentially infectious material.
Table of Contents Add to my manuals Add. Get the results in a matter of seconds and minutes: After 2 minutes with test strip inserted; after 1 minute if no button was pressed and no test strip has been inserted. The Accutrend GCT System is most suitable for accurate on-site screening, diagnosis, and therapy monitoring of metabolic disorders and cardiovascular risk factors.
Accutrend GCT System is a combo monitoring unit for cholesterol, blood glucose and triglycerides, all in a fast, easy and reliable measurement process. View digital manual for this product.
Then CODE stops flashing. If maunal strip code has not been read in correctly, error messages appear e. Don’t show me this message again.
Code the meter see Section 3. Move the meter to a dry environment and wait a while before switching it on again. The hours flash on the display.
System Components For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office. When all positions have been shown the display accutrens to To count down from any value stored: Plausibility check 37, 69 Practising measurement 29ff, 59 Retrieval of results from memory 49ff. Ready for setting the date and time or the beeper. Check the expiry date and storage accuhrend of the test strips.
It should also be cleaned: Safety warnings such as are also stored. If xccutrend of these items are missing or damaged please contact your supplier. Illustrations of the meter are shown on the fold-out page. Try a Test-drive for Yourself Now!
Make sure that the strip is inserted as far as it will go and accutredn the word GLU on the strip can no longer be seen. Light from an LED light emitting diode is directed towards the test strip from below and the blank value of the test strip determined by measurement of the reflected light.
Roche Accutrend GC User Manual
The relevant regulations or guidelines applying to your institute or laboratory should be observed. Press the SET button briefly. The GCT System, while greatly suited for clinical testing, is also personal testing made easy. The Accuchek GCT System lets you keep track of the trend by using its memory function to store your measurements, up to 50 glucose, 20 cholesterol and 50 triglyceride values. With the GCT System, you easily do these tests on your own, at your convenience, with accurate, dependable results.
Troubleshooting Move the meter to a dry environment and wait a while before switching it on again. Free Health Articles Select a topic to access it.
ROCHE ACCUTREND GC USER MANUAL Pdf Download.
If you suspect a fault in your meter you should first contact the advisory service at your nearest Roche Diagnostics office. The gc strip is too old or has been incorrectly stored. If the button is released earlier the clearing process will be interrupted. Keep pressing to set the minutes in the same way as the hour.
They are essential to ensure precision and reliability of the measurements. The blood must be applied to the test strips outside the meter. Contact the Roche Diagnostics service centre. Speedy as when when you delivered to me in Kaduna.